1a. Pulsator Test Single App
The Pulsator Test Single App records Pulsators. A minimum of 6 cycles will be recorded and analyzed in order to calculate to different phases according to ISO 6690 and NMC norms. Each measurement can be identified by entering an identifier-name, also a note to each measurement can be entered afterwards before saving the measurement to the memory of the PT-VI.
1b. Pulsator Test Sequence App
The Pulsator Test Sequence App records Pulsators in a way that the user can enter an identifier-name manually before recording an series of pulsators in sequence order . A minimum of 6 cycles will be recorded and analyzed in order to calculate to different phases according to ISO 6690 and NMC norms. A note can also be entered before saving the measurement. The tester stays in the measurement menu and will increase the identifier with one position and will repeat this procedure until the whole series of pulsators has been recorded.
1c. Constant Vacuum (Vacuum Fluctuations) App
The Constant Vacuum App records vacuum fluctuation in an undefined time frame. The sampling rate can be set by the user before starting the recording. The PT-VI records data until the user stops the recording manually. From the recorded data the minimum, maximum and average value will be calculated and the results can be saved with an identifier and notes. Recorded data can be zoomed in parts and from each part that has been zoomed the minimum, maximum and average data will be recalculated and can be “saved as” with a new identifier and notes. Constant Vacuum recordings can also be recollected from the memory afterwards and from there resaved as parts of it. During recording of data a marker can be added at a moment when an event occur. Event description of a marker can be written in the notes of each recording or part of it.
1d. Vacuum Gauge App
The Vacuum Gauge App measures the connected vacuum level and gives the value as a static digit in the screen. Each measurement can be provided and saved with an identifier and notes.